Durr CS 1 Combi-Sepamatic
Membership is strictly limited to registered dentists in Malaysia only
The CS 1 has a powerful drive so that the suction system can cope even with treatment with a high fluid flow rate.
The two-stage separation prevents blood foam being drawn in and thus protects the connected dry suction machine.
Cleaning and disinfecting the suction unit
- After every patient treatment the equivalent of one glass of cold water should be sucked though the system, i.e. through both the large and the smaller suction hoses – even when, during treatment, only the saliva extractor has been used.
- Before any midday break and at the end of the working day the suction unit should be cleaned and disinfected by suction of a nonaggressive, non-foaming cleaning and disinfectant agent.